Bacteria in dentine infected by caries leave behing metabolic products (Porphyrins).
When an exposed cavity is illuminated with violet light, these porphyrins show a red fluorescence, thereby indicating the significant areas for caries excavation.
This caries detector aids detection and monitoring of carious tooth substance, in particular in dentin, during the excavation of previously opened cavities.
Accoring to the particular fluorescence properties of the degradation product which is produced by the cariogenic bacteria, substances with teeth for removing infected, to remove the infected material on the teeth.
1.Easy to use
Dentine infected with caries is clearly visible in opened cavities when illuminated with the caries -detected light.
The dentine infected with caries present there fluoresce red,which healthy tooth substance fluoresces green.
The disgnostic goggles increase the contrast.
2.More reliable caries detection
An examination with Caries -detected light minimizes the risk of recurrence of caries.
It also provides important information for deciding how to treat Caries Profunda.
3.High efficiency with caries excavation
You assess the success of caries excavation at a glance at any time during the process.
4.Supports minimally invasive procedures
You can precisely excavate dentine infected with caries.This conserves the healthy tooth substance of your patients!
Effective function for decayed tooth- detection , quickly, easily, painless.
Wireless operation, Lightness using,easy to use;
Support minimally invasive surgery, safely, lighten patient's fear.
Detachable rechargeable battery.
Over 80 minute of continuous use on a single charge.
Tow type optic fibers (90°or 140°).
The light rod can be sterilized under the temperature up to 135 ℃ (274 ℉).
Adaptor working voltage: AV100-240V, 50-60Hz
LED intensities: 60-250mw
LED wavelength: 380-420nm
Diagnostic goggles: >500nm
1 X Base body
2 X Light rod
1 X Battery
1 X Diagnostic goggles
1 X Charger Station
1 X Adapter
How to charge?
1 Remove the discharged battery from the light probe.
2 Insert the charged battery into the light probe unit it screws tight into place.
3 Position the discharged battery onto the charge station.
4 Press the button on the battery.As soon as the charging light switches on,the battery is been charged